cSAXS – X12SA: Coherent Small-Angle X-ray Scattering
cSAXS is a beamline dedicated to small-angle X-ray scattering (SAXS) and coherent diffractive imaging, in particular ptychography. Located at the Swiss Light Source at the Paul Scherrer Institut in Villigen, Switzerland. For information on experimental setups, geometries, and some sample preparation details see the beamline's endstations website. We strongly encourage all prospective users to discuss their experimental plans prior to proposal submission with the beamline staff.
Access to cSAXS is granted through the biannual SLS call for proposals. Alternatively access can also be granted through the European softmatter infrastructure E(U)SMI.
Access to PSI data analysis center is described in Photon Science Data Analysis.
We have a Postdoctoral Fellow position currently open. The project is a collaboration between Chalmers University of Technology (Goteborg, Sweden), Paul Scherrer Institut (Villigen, Switzerland), and MAX IV (Lund, Sweden). The employment contract at PSI is limited to 1.5 years, after this period another contract for 1.5 years will be available by MAX IV.
Experiments performed at cSAXS include, inter alia,
- Ptychographic nanotomography, see for example Dierolf et al., Nature 467, 436 (2010). Dedicated instrumentation, which is available to users, is described in Holler et al., Sci. Rep. 4, 3857 (2014).
- Optics characterization, see for example Vila-Comamala et al. Opt Express 19, 21333 (2011)
- Small-angle X-ray scattering (SAXS), e.g., Smeulders et al., Nature 478, 412 (2011), can be performed with spatial resolution as microscopy technique in 2D (Bunk et al. New J. Phys. 11, 123016 (2009)) and 3D (Liebi et al. Nature 527, 349 (2015)). SAXS is also available with time resolution in the millisecond range (Westenhoff et al., Nature Methods 7, 775 (2010)).
|Energy range||4.4 – 17.9 keV|
|Energy resolution||ΔE/E < 0.02%|
|Spot size on sample (horizontal × vertical)|| variable from 1.5 × 0.8mm2 to 25 × 10µm2.
Smaller beam sizes are possible using Fresnel zoneplate lenses, for more details contact the beamline staff
|total flux (at sample position)*|| ~2x1012photons/sec (at E = 6.2keV, 400mA ring current)
~1x1012photons/sec (at E = 11.2keV, 400mA ring current)
~5x1011photons/sec (at E = 13.6keV, 400mA ring current)
|Coherent flux**||~7×108photons/sec (at E = 6.2keV, 400mA ring current)|
|q-range (@11.2 keV)†|| SAXS detector at 7.2 m, q = [0.0012, 0.1]Å-1, d-spacing = [524, 6.28] nm.
SAXS detector at 2.1 m, q = [0.004, 0.34]Å-1, d-spacing = [157, 1.85] nm.
WAXS detector [0.64, 2.8]Å-1, d-spacing = [1, 0.22] nm (vertically oriented strip detector, no orientation information from anisotropic scattering is available for the WAXS detector).
**at the sample position in a beam focused by a Fresnel zoneplate with 60 nm outermost zonewidth and approximate beam waist diameter of 100 nm.
† the lower ends of the q-range for the SAXS detector require a special beamstop arrangement and are significantly more affected by parasitic scattering. If you need very low q-range information, below e.g. 0.002 Å-1, please discuss your experiment ahead of time with the beamline staff.